Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||13 May 2008|
|PDF File Size:||5.93 Mb|
|ePub File Size:||11.16 Mb|
|Price:||Free* [*Free Regsitration Required]|
John Fluke Model 8040A Digital Multimeter
See other items More 804a0 Designation or Item Number. Maintenance is minimized by encapsulation of the divider components. Disconnect the input from the V-H input. An 01 Option is provided for flukke having an 11 megohm input resistance, and an 02 Option is provided for voltmeters having an infinite input resistance 10 3 megohms or 8040 at null.
All Items will be shipped out within 48 hours of purchase if it is a item that must be palletized I need a commercial address and phone number. The figures, presented here in the trouble- shooting section, are of signal waveforms taken at selected points in the A circuitry. Avoid moving the componentsespecially after the calibration procedure has been completed.
Probable cause, U25 defective. Subject to credit approval. Autoranging or Manual Accuracy: Select the desired current range flume accordance with Table The value of Vref is determined by algebraically adding Vx a positive voltage to a negative equivalent of the ohms source voltage. Contact the seller – opens in a new window or tab and request a shipping method to your location.
These solutions will react with the plastic materials of the instrument. Most corrosive agents will not damage the probe body.
The A will display the temperature in kilodegrees. RF signals applied to the 80T probe tip can also cause errors in temperature measurement.
Any international shipping and import charges are paid in part to Pitney Bowes Inc. Rechargeable Ni-Cad pack 8 hr.
See Appendix A for Code-to-Name list.
The battery can be normal cell or rechargeable Ni-Cd. A mV and 2V ranges: Figure provides a graphic illustration of switch positions and input connections for each A function. The peg, when the battery pack is in 804a on the A, closes the charging path allowing the batteries to be recharged.
Antique Radio Forums • View topic – WTB Fluke a Digital Multimeter Vintage Desktop Model
Ohms Voltage Source If this is true yet the amplitude of the second segment of the waveform, Operating Notes Block diagrams and simplified circuit discuss the operation of the A in terms of the two diagrams are included, where needed, to aid in major sections and subsections of each giving a brief understanding the theory.
There are 1 items available. When replacement is necessary, use a 2 amp AGX replacement fuse. Add to watch list Remove from watch list. Analog Circuits This amount is subject to change until you make payment.
Full text of “FLUKE a”
To calibrate the Model 80F-5 perform the following steps: Select the AC V function and the 2 range. Unscrew the strain relief on the input cable of the Model 80F-5 and remove the front cover.
For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees. Not enough time to further test the unit. All digits are blank.